Place the device over the palm of your hand (photo 2) and press the Start/Stop button to power the device. will run a self-test* (3 lights will blink).When 2 lights will turn green is ready for use.
Press the Mode button to select treatment program.
Press the Start/Stop button to begin treatment. The treatment light will begin to blink. Hold the device over the area being treated for the duration of the treatment.
In conclusion of the treatment, the treatment light will stop blinking and a short beep will be heard. Press the Start/Stop button to repeat the treatment It is possible to stop a treatment at any time by pressing the Start/Stop button Turn off by pressing on the Start/Stop button for 3 seconds.
*Important: is programmed to run a self-test every 24 hours.
When you do not turn on the device over the palm of your hand, the device beeps and 4 lights begin to blink. Please turn off the device and power it again over the palm of your hand.
The user can choose between 3 programs which have been pre-set to treat different pain conditions. A program is selected by pushing the mode button.
Each program uses different pulse frequencies, as displayed in the table below. Pulse frequencies variations alter the laser light’s depth of penetration.
Where possible, treatment should be provided over bare skin. During treatment, the device should be held in a hovering state over the area where the pain is felt. If the treatment area is wider, the device should be moved (scanned) around the treatment area.
If required, treatment can be provided over clothing or over wound dressing.
We strongly recommend you use the device in strict compliance with the safety pre-cautions and operating instructions in this manual.
Handycures' should be maintained and serviced by Medical Quant Limited personnel or other qualified personnel approved in writing by Medical Quant Limited. Please review the manual before initial use. It is important to keep this manual handy.
|Battery light is red
|| Handycures' rechargeable battery has run out of power
||Connect Handycures' to the electricity until the battery light turns green Handycures' may be operated while recharging
|All 3 program lights are showing
||Wrong mode of device operation||Power the device as specified in(operating instructions)|
|Technical fault||Contact the dealer|
Note: When in operation, only the red light is visible to the eye. If no red light lights on, contact the dealer.